9V ASI battery is located in the battery pack in the 9V battery
compartment (see below). To install, remove the cover covering the 9V
battery compartment by pushing on it sideways. The cover will slide
approximately a 1/4 inch and then can be detached from the battery
pack. Insert the 9V battery into the 9V battery compartment so that
the contacts on the battery touch the contacts in the battery pack.
The orientation of the battery contacts is shown in a picture on the
inside bottom of the 9V battery compartment. Replace the 9V battery compartment door by reversing the process used to remove the door. We recommend only using 9 Volt lithium batteries for the Defibtech AED.
Note: 9 Volt
Lithium battery only powers self test procedures in the Defibtech
Lifeline and Reviver AED's. 9-volts battery should be replaced at least annually
into Defibtech Battery Packs. DAC-410
We recommend that the Defibtech Lifeline AED be inspected at least monthly to assure rescue readiness. Use the checklist below to inspect your AED. For further information or for troubleshooting refer to the Defibtech Operations Manual.